MiniMed 670G System FAQs
MiniMed™ 670G system
Frequently asked questions (FAQ)
NEED HELP? SEARCH AN FAQ TOPIC.
Three daily tips can help you take care of your insulin pump so it takes care of you!
- Use a pump case: Silicone pump cases can help provide a cushion against bumps during your daily activities.
- Keep it secure: If your pump touches your skin, turn the buttons away from your body to reduce long term moisture exposure. Consider a waist pouch while exercising.
- Soft button pushing: When pushing the buttons on your pump, use the side or pad of your finger. Try to avoid sharp objects like keys or the tips of long or acrylic fingernails.
The temporary basal rate feature, for use in Manual Mode, lets you immediately increase or decrease your basal insulin for the period of time (duration) that you set. It is often used for exercise and sick days. A Temp Basal can be set in either percent (delivers a percent of the current basal rate) or by rate (delivers the amount that you enter). Follow these pump screen instructions.
While in Auto Mode, you may set a temporary sensor glucose (SG) target (Temp Target) of 150 mg/dL for situations in which you would like your target to be temporarily higher, such as exercise. Check with your healthcare professional regarding use of a Temp Target.
- Try not to calibrate with multiple blood glucose (BG) meters. It’s best to stick with the same meter for all your calibrations.
- Wash your hands before checking your BG.
- Enter the BG reading into the pump right away. Don’t use old meter BG readings or previous sensor readings as a calibration.
- If your system requires a new calibration, wait at least 15 minutes between calibration entries.
You should calibrate when your glucose levels are stable. Exercise and eating can cause rapid changes in your glucose levels. That’s why many people choose to calibrate before meals and before bedtime. It’s important to always calibrate when the system asks you to or you will be exited from Auto Mode. For more information, take a look at this video about calibration best practices.
If you receive a “Calibration not accepted” alert, it means that the pump couldn’t update your sensor using the blood glucose (BG) value you entered. This could be because the BG entered was not within a range the system was anticipating. It may also be caused by rapidly changing glucose levels, delayed or incorrect BG entries, or it may be an indication the sensor is not working properly. If you receive a “Calibration not accepted” alert, wait approximately one hour (or until your BG is stable) and then try to calibrate. If you have recently eaten, exercised, or taken insulin, you might need to wait a little longer than an hour.
You can use the Calibration Reminder to give you notice before the next calibration is necessary. For example, if you calibrated at 7:00AM your next calibration would be due at 7:00PM (12 hours later). If your reminder is set for 1 hour, you would receive a Calibration Reminder at 6:00PM which is 1 hour before the calibration is due. This can help ensure you calibrate 3-4 times a day. The Calibration Reminder defaults On with a reminder time of 1:00 hour.
To change the Calibration Reminder:
- Press Select > select Options > select Reminders > select Calibration.
- Select Reminder to turn the reminder on or off.
- If you turn on the reminder, select Time and enter a time between 5 minutes and 6 hours.
- Select Save.
CGM alerts & alarms
CGM alerts at night may be caused by a few different reasons ranging from sensor insertion to site placement. Here are a few tips to help prevent getting CGM alerts in the middle of the night.
- Sensor insertion, site placement and taping make a difference. For your sensor to work its best, avoid sites where clothing may rub or constrict, where your body naturally bends which may cause the sensor to pull out and where this a lot of pressure. Follow instructions for sensor insertion and taping here.
- Check your pump’s home screen before going to bed. If you’re using the SmartGuard™ Auto Mode feature, make sure you see the blue shield. Check your blood glucose and give a correction bolus if one is suggested.
- Know your calibration status. The calibration icon shows how much time is left until your next scheduled calibration is due, and calibrating before bed helps reduce the chance of needing a calibration overnight while sleeping.
- Customize nighttime alert settings. Talk to your healthcare team about high and low glucose limits overnight to make sure alerts are personalized to meet your needs. You may consider setting both high low limits differently for the overnight period.
- Plan ahead. Sleeping in on the weekend? Summer schedule different from the rest of the year? If your schedule changes, think about how your calibration schedule may need to be adjusted to help avoid any unwanted alerts and alarms.
- Consider the Alert Silence feature. Your pump has a feature called Alert Silence, where you can temporarily silence alerts. Don’t worry, you will still get important low alerts if Alert Silence is ON. For your safety, Low SG (50 mg/dL or below) will always sound, and Alert on Low will sound if it’s turned on. Learn how to set Alert Silence here.
A few things might have happened:
- You might have received two consecutive “Calibration not accepted” notifications
- You might have accidently selected “No” on the “Check sensor insertion—Is sensor fully inserted?” screen
- The sensor might not be working properly and needs to be replaced
For all three scenarios, you will have to change your sensor.
Check out this Sensor Alerts Quick Reference for the most common alerts, reason they occur and steps to take.
Your pump has a feature called Alert Silence, where you can temporarily silence your alerts. To use the Alert Silence feature:
- Go to: Main Menu > Audio Options > Alert Silence Options
- Select All Sensor Alerts or your desired alerts
- Select the Duration (time) you want, and select Begin
Don’t worry, you will still get important low alerts if Alert Silence is ON. For your safety, Low SG (50 mg/dL or below) will always sound, and Alert on Low will sound if turned on.
You can cancel the Alert Silence feature at any time:
- Go to: Main Menu > Audio Options > Alert Silence
- Select Cancel Alert Silence
Taping & securing your sensor
The MiniMed™ 670G system uses the Guardian™ Sensor 3. For those ages 14 and up, the sensor can be worn on the back of the upper arm or your abdomen. For those ages 7-13, the sensor can be worn on the abdomen or buttock.
We have a recommended method for taping that we have found is effective for most people who wear CGM. It requires you to use both pieces of oval tape that are supplied with your sensor. Some people also use an additional layer of tape called overtape. Watch the video below on how to tape and use extra adhesives.
Here are some things you can try to help get up to 7 days wear out of your glucose sensor:
- Many people have success wearing the sensor on the back of the upper arm.
- Follow our taping instructions. Incorrect taping is one of the most common reasons why CGM sensors won’t stay in place.
- Try using a liquid adhesive to help your tape stick for longer. Make sure to apply the extra adhesive after you insert your sensor and before you put your tape on.
Watch the video below to see what other people who wear the system recommend.
Check out this informational guide to help with additional site management. Check with your healthcare team about solutions that may be best for you.
Contact 24-Hour Technical Support at 1-800-646-4633 option 1 for technical assistance or to report product issues. They can also provide a sample kit with various tapes and adhesives.
Requesting a replacement sensor
A BG required alert is required to stay in Auto Mode. Calibrations are to ensure you continue to receive your readings, and to make sure those readings are accurate.
If you happen to get a BG required alert, and it is also time to calibrate, go ahead and use the blood glucose for a calibration. But you do not have to calibrate every time Auto Mode asks you for a BG. When might you not want to calibrate? If your glucose levels were changing rapidly, say you saw 2 or 3 arrows on the screen, you may not want to calibrate.
If you receive a Cal Not Accepted, look at your pump screen to confirm there is no Sensor Updating alert and wait at least 15 minutes before entering a new blood glucose (BG). Some people find that waiting longer, like 1 hour, helps.
Always use a new BG after getting this alert, and make sure you have clean hands. After a 2nd consecutive Cal Not Accepted alert, you’ll get a Change Sensor alert, and we want to avoid this when possible.
Auto Mode checks each blood glucose (BG) that is entered, and requirements are stricter than those for CGM alone.
If you get a BG required after you’ve just entered one, it means the first BG was not accepted by Auto Mode and the system requires additional BGs. If you did not calibrate on the first BG request, you can use your BG to calibrate your sensor. It may help to wait 30-60 minutes before re-checking and entering another BG or calibration.
You may receive additional requests to Calibrate now, especially on day 1 of wearing a new sensor. This is normal, and it’s your pump’s way of making sure your sensor is performing well. So spread your calibrations throughout the day. Calibrate your sensor with your pre-meal BGs as well as before bed to help avoid nighttime alerts.
Sensor updating alert means your pump is performing sensor checks. In other words, your system is deciding if your sensor glucose values are reliable.
There are different reasons why you might see “Sensor updating”. Your pump will let you know that it could last up to 3 hours, but most times it will resolve sooner than that. Allow your pump to perform its checks, and simply follow any instructions on your pump screen.
If you find that you see ‘Sensor updating’ often, and then you see ‘Change your sensor,’ you may need to check your sensor insertion sites, perhaps rotate locations, and make sure you’re taping your sensor down well. Get help if you have questions by reaching out to the 24-Hour Technical Support or your Medtronic representative.
Even if you have an online Medtronic account, a unique login is needed for CareLink™ software. Learn how to create a CareLink account.
The CareLink™ uploader allows you to upload the information in your pump or meter to the CareLink™ website. Learn how to set up CareLink™ on your computer.
Check out the Upload Guide for easy steps for your first or next pump upload.
The Assessment and Progress Report is designed to help you view your glucose management while on your MiniMed™ 670G system. You can use this report with your healthcare professional to improve the duration of time spent in Auto Mode and determine what events caused some of your Auto Mode exits. Learn more about the Assessment and Progress Report.
Tips for traveling
Pack extras of everything– insulin, infusion sets, sensors, reservoirs, alcohol swabs, sensor tape, extra compatible batteries for your pump, and even syringes in cases of emergency. The mentality is, “it’s better to have more of what you need, than not enough.” Print out the travel checklist to keep handy.
The TSA allows for diabetes-related supplies, equipment and medication—including liquids—through the checkpoint once they have been properly screened by X-ray or hand inspection.
If possible, pack all your supplies together in your carry-on bag so you have everything on hand. Before your screening begins, inform the officer conducting the screening about any supplies on you or in your carry-on.
Read more about flying with your diabetes supplies at how-to pass-through security before boarding your flight.
The Travel Loaner Program allows US-based customers to take a "back-up" insulin pump for a $50 fee every 90 days. Take a "back-up" pump when you:
- Go on a cruise
- Travel to Hawaii or Alaska
- Travel internationally
Complete the Travel Loaner form and submit it at least 2 weeks prior to your trip to ensure on-time delivery. A signature is required upon delivery and you will need to return the travel loaner when you arrive back home.
It is important that you check your blood glucose (BG) more frequently while you are traveling. The routine hassle of travel, including stress, changes in time zones, schedules and activity levels, meal times and types of food, can all affect your diabetes control. Be extra attentive to monitoring your BG frequently and be prepared to respond if needed.
Note: The Travel Loaner Program only includes insulin pumps and does not include blood glucose meters or CGM devices.
Tips for Exercise
Remember, the standard Auto Mode target is 120 mg/dL, and you can temporarily change your target to 150 mg/dL. The Temp Target feature is designed for any time you want less background insulin. In addition to exercise, this may include yard work, times of high stress, exams, driving, or drinking alcohol. Every person is different, but in general you could start the Temp Target 1‐2 hours before exercise and end it 1‐2 hours after.
If you find you are still dropping low during exercise, you may need to consider a combination of suspending your pump and/or consuming carbs to avoid the low altogether.
You might also benefit from these additional these additional exercise tips. Check out Exercise Strategies for People with the MiniMed™ 670G System on the LOOP Blog.
We recommend ordering when you open your last box of sensors or infusion sets. To order supplies, you have three options:
- Sign up for Scheduled Orders or place your one-time order at www.Diabetes.Shop.
- Text SUPPLY to 22094 to start the order process.
- Call us between 8AM-6PM CT Monday - Friday at 1-800-646-4633, Option 2.
If you order sensors through a distributor, you’ll need to contact them directly to place your next order.
For all things finances, from how to pay your bill to options for financial assistance, visit this page.
Have More Questions?
Search for additional support.
1-ON-1 Diabetes Care Specialist Support
Schedule a 1-on-1 call with a Certified Product Trainer.
Free MiniMed™ 670G Live Webinars
Sign up today for a webinar and gain insights into Auto Mode.
MiniMed™ 670G Product Support
View product operating instructions and helpful support videos.
Important Safety Information: MiniMed™ 670G System with SmartGuard™ technology
The Medtronic MiniMed™ 670G system is intended for continuous delivery of basal insulin (at user selectable rates) and administration of insulin boluses (in user selectable amounts) for the management of type 1 diabetes mellitus in persons, seven years of age and older, requiring insulin as well as for the continuous monitoring and trending of glucose levels in the fluid under the skin. The MiniMed™ 670G system includes SmartGuard™ technology, which can be programmed to automatically adjust delivery of basal insulin based on Continuous Glucose Monitor sensor glucose values and can suspend delivery of insulin when the sensor glucose value falls below or is predicted to fall below predefined threshold values. The system requires a prescription. The Guardian™ Sensor (3) glucose values are not intended to be used directly for making therapy adjustments, but rather to provide an indication of when a fingerstick may be required. A confirmatory finger stick test via the CONTOUR®NEXT LINK 2.4 blood glucose meter is required prior to making adjustments to diabetes therapy. All therapy adjustments should be based on measurements obtained using the CONTOUR®NEXT LINK 2.4 blood glucose meter and not on values provided by the Guardian™ Sensor (3). Always check the pump display to ensure the glucose result shown agrees with the glucose results shown on the CONTOUR®NEXT LINK 2.4 blood glucose meter. Do not calibrate your CGM device or calculate a bolus using a blood glucose meter result taken from an Alternative Site (palm) or from a control solution test. It is not recommended to calibrate your CGM device when sensor or blood glucose values are changing rapidly, e.g., following a meal or physical exercise. If a control solution test is out of range, please note that the result may be transmitted to your pump when in the “Always” send mode.
WARNING: Medtronic performed an evaluation of the MiniMed™ 670G system and determined that it may not be safe for use in children under the age of 7 because of the way that the system is designed and the daily insulin requirements. Therefore this device should not be used in anyone under the age of 7 years old. This device should also not be used in patients who require less than a total daily insulin dose of 8 units per day because the device requires a minimum of 8 units per day to operate safely.
Pump therapy is not recommended for people whose vision or hearing does not allow recognition of pump signals and alarms. Pump therapy is not recommended for people who are unwilling or unable to maintain contact with their healthcare professional. The safety of the MiniMed™ 670G system has not been studied in pregnant women. For complete details of the system, including product and important safety information such as indications, contraindications, warnings and precautions associated with system and its components, please consult http://www.medtronicdiabetes.com/important-safety-information#minimed-670g and the appropriate user guide at http://www.medtronicdiabetes.com/download-library
© Medtronic. All rights reserved. Medtronic, Medtronic logo and Further, Together are trademarks of Medtronic. Third party brands are trademarks of their respective owners. All other brands are trademarks of a Medtronic company.
Ascensia, the Ascensia Diabetes Care logo, and Contour are trademarks and/or registered trademarks of Ascensia Diabetes Care.