Manage Settings Error
Delivery stopped. Backup settings cleared from Manage Settings. Current settings are working properly. Select OK to restart. See User Guide.
What has happened: A pump error has occurred, and you need to restart your pump. Your backup settings have been lost, but your current settings are unchanged.
- Select OK to restart your pump. Your current settings are unchanged. Only your backup settings are lost
- When the pump restarts, follow instructions on the pump display
- If the pump was delivering a bolus or filling the cannula, check Daily History and evaluate your need for insulin
- Consider saving your current settings