Returning to SmartGuard™ feature
Device: MiniMed™ 780G insulin pump
The SmartGuard feature may stop functioning under the following conditions:
The SmartGuard feature is turned off.
The pump is delivering basal insulin based on insulin delivery history, and not SG readings, for four hours. See staying in the SmartGuard feature for more information.
All insulin delivery has been manually suspended and has not resumed for four hours.
The Sensor feature is turned off or the transmitter is disconnected.
Returning to the SmartGuard feature after an exit:
The pump indicates any required actions on the Home screen after an exit from the SmartGuard feature. In the example below, a BG entry is needed. Once the BG is entered, the pump resumes using the SmartGuard feature.
System support User guides and manuals