MiniMed™ 770G System Support
Stopping and Resuming Insulin Delivery
Device: MiniMed™ 770G Insulin Pump
Use Suspend Delivery if you need to stop all active basal and bolus insulin deliveries. While your insulin delivery is suspended, your pump beeps, vibrates, or both depending on your audio settings. This occurs every 15 minutes to remind you that insulin is not being delivered.
To suspend all insulin delivery:
Press select and select Suspend Delivery.
To suspend press Right and select Yes. The Delivery Suspended Successfully confirmation will appear.
After suspending delivery, the home screen will show your insulin delivery is suspended. Your pump functions are limited until you resume your basal insulin delivery.
To continue your basal insulin delivery, use the Resume Delivery feature. Your pump starts your programmed basal pattern but does not start any previously programmed bolus deliveries.
To resume basal insulin delivery:
Press select and select Resume Delivery.
You will be asked to confirm resuming basal delivery. Press right and select Yes. Delivery Resumed Successfully will appear on the screen.