Before Using SmartGuard™ Auto Mode
Device: MiniMed™ 770G Insulin Pump
SmartGuard™ Auto Mode can be enabled at any time, but it does not activate until the system completes a 48-hour warm-up period while you use the pump to deliver insulin. This warm-up period begins the midnight after the pump starts delivering insulin and it does not require sensor use. During the warm-up period, your Auto Mode system collects and processes data that help enable its automatic function.
To prepare your pump for SmartGuard™ Auto Mode:
- Cancel any active Temp Basal rates.
- Make sure your insulin delivery is not suspended on your pump.
- Set your carb ratio.
- Review your high and low limit settings. Your high and low limit settings apply to Auto Mode.
- Enter a BG reading if you have not entered one in the past 12 minutes. If necessary, calibrate your sensor. If you have just started a new sensor, calibrate your sensor, and then wait 30 minutes before you enter a BG for Auto Mode.