MiniMed™ 770G System Support
Adding a New a Basal Pattern
Device: MiniMed™ 770G Insulin Pump
To add a new basal pattern:
From the Home screen, press Select.
Select Delivery Settings.
Select Basal Pattern Setup.
To add a new basal pattern, select Add New.
Select a basal pattern name.
Press Select on the time segment.
To set one continuous 24 hour basal rate, leave the end time at 12:00A by pressing Select. To set multiple basal rates, set the end time for the first rate using up and down buttons and press Select again.
Use the up and down buttons to set your rate in units per hour. Press Select.
Continue to set rates for the different time periods as needed. The End time for your last rate must be set to 12:00A. Press Select after each entry.
Review basal rates. The End time of the final segment must be 12:00A to complete the pattern.