MiniMed™ 770G System Support
Setting Up the Easy Bolus™ feature and Delivering a Bolus
Device: MiniMed™ 770G Insulin Pump
Before you use the Easy Bolus™ feature, you must turn on the feature and set the step size. The step size determines the number of units the bolus amount increases each time you press the Up button. Your Easy Bolus™ delivery is limited to 20 steps or your max bolus limit, whichever comes first.
Easy Bolus™ is not available while in SmartGuard™ Auto Mode.
To set up the Easy Bolus™ feature:
Select Delivery Settings.
Select Easy Bolus.
Select Easy Bolus again to turn it on.
Set the Step Size amount in units. You can set the step size from 0.1 to 2.0 units. Your step size cannot be higher than your max bolus.
To use the Easy Bolus™ feature, your pump must be in sleep mode. Your pump automatically goes into sleep mode two minutes after the screen turns off.
To deliver a bolus using the Easy Bolus™ feature:
While your pump is in sleep mode, press and hold Up for about one second. After your pump beeps or vibrates, release. You can now start to program your bolus with the Easy Bolus™ feature.
Press the Up button the number of times needed to set your bolus amount. Each time you press Up, your pump makes a tone or vibrates, and your bolus amount increases by the number of units set for the step size.
When you reach the desired bolus amount, press and hold Up to confirm the amount. Your pump beeps or vibrates for each button press. Count to ensure the amount is correct. If the amount is incorrect, press and hold Down until you hear a tone, and then start again from step 1.
When the bolus amount is confirmed, press and hold Up for about one second to deliver your bolus. Your pump beeps or vibrates. Your bolus starts immediately after the confirmation.